I have bought some LL1673/10H from you, but the measured inductance (with my inductance meter) is only 6H. What is wrong?
Our power supply chokes are designed to work at 100/120Hz and with voltages around 100V. Most induction meters work at a few volts or less, at around 1kHz. This is not the correct operating point for a power supply choke, and the reading on the induction meter is usually around half the specified value of [...]